"TrList.ini" installation instructions
Transducer definition file upgrade. This page describes how to upgrade the TrList.ini file.
The TrList.ini file contains the transducer parameters obtained during calibration. It also contains generic parameters related to transducer settings and limitations.
When the TrList.ini file has been replaced, the transducer calibration data is lost.
Download the TrList.ini file.
Close down the ER60 echo sounder application, and access the computer's file structure.
Locate the current TrList.ini. By default, the file is located in directory
Windows XP: C:\Program Files\Simrad\Scientific\EK60
Windows 7: C:\ProgramData\Simrad\ER60\ConfigSettings
Make a backup copy of the current file.
Copy the new TrList.ini into the same directory.
Restart the ER60 echo sounder application.
Calibrate the echo sounder system before you use it for scientific purposes.
08 April 2013
|Released with new SW version 2.4.3 for the Simrad EK60.
24 May 2012
|New transducer added:
10 July 2009
|New transducers added:
Note that only the latest version of the TrList.ini file can be downloaded.